Der CS705 ist ein Schneefall-Adapter, der eine Regenkippwaage mit 200mm Durchmesser ganzjährig verwendbar macht. Normalerweise wird er mit der TE525WS eingesetzt.Lesen Sie mehr
The CS705 consists of an antifreeze reservoir, overflow tube, and catch tube. Snow captured in the catch tube dissolves into the antifreeze. The melted snow raises the level of the antifreeze and water solution. The mixture flows through the over-flow tube into the tipping bucket where it is measured by the tipping bucket mechanism.
The CS705 possesses inherent delays and is not suitable for real-time precipitation measurements. The following factors contribute to the delays:
For rainfall at 25°C, a delay of minutes is expected after the gauge receives a minimum accumulation of ~0.03 inches. For snowfall, a delay of hours to tens of hours is expected. The longest delays should be expected for low density snows at very cold air temperatures. However, all precipitation falling into the catch tube eventually flows through the overflow tube and is measured by the tipping bucket rain gauge.
|Recommended Antifreeze Type||
1:1 mixture of propylene glycol and ethanol (PGE)
Campbell Scientific does not recommend using the antifreeze that is typically used in cars or RVs; refer to the Compatibility section for more information.
|Capacity||20.3 cm (8 in.) of liquid @ -20°C (assuming 1:0 starting ratio of antifreeze to water)|
|Antifreeze Reservoir Capacity||9.46 L (2.5 gal)|
|Antifreeze Reservoir Diameter||20.96 cm (8.25 in.)|
|Antifreeze Reservoir Height||35.6 cm (14 in.)|
|Catch Tube Orifice||20.3 cm (8 in.)|
|Catch Tube Height||25.4 cm (10 in.)|
Tipping Bucket Rain Gages
The CS705’s specially shaped cylinder allows it to mount to the 8-in. orifice of the TE525WS Tipping Bucket Rain Gage. The CS705 cannot be directly used with the TE525, TE525MM, CS700, TB4, or TB4MM tipping bucket rain gages. Both the TE525 and TE525MM can be converted to a TE525WS by returning them to Campbell Scientific.
The CM270 Mounting Kit is included with each purchase of the CS705 Snowfall Adapter. The kit consists of a funnel, two band clamps, and mounting hardware. The funnel drains the antifreeze/precipitation mixture into a pail. One of the band clamps secures the top of the CS705 to a pole. The other band clamp attaches the funnel and the bottom of the rain gage to the pole.
A user-supplied pole or a CM300-series mounting pole may be used. The user must provide a pail and a hose with a 0.5 in. inner diameter. The hose must be long enough to span the distance from the funnel outlet to the pail.
Number of FAQs related to CS705: 5
Alle anzeigenWenige anzeigen
Yes. In the Quickstart section of the instruction manual, Figure 4-1 and Figure 4-2 provide labeled images of the parts with their corresponding part numbers.
Yes. Mixed rain and snow does not cause any issues for the CS705.
The CS705 works with the TE525WS-L rain gage.
The CS705 Tipping Bucket Snowfall Adapter cannot be used directly with the TE525-L or TE525MM-L because of the funnel size. To make the rain gages compatible, they can either be sent to Campbell Scientific to be converted to a TE525WS-L, or an 8 inch funnel can be installed. If the funnel is used, see the instruction manual for the appropriate multipliers, as they will change.
Generally, the antifreeze is changed after 8 inches of liquid water have been added to the antifreeze/ethanol mixture. Eight inches of water will dilute the antifreeze/ethanol mixture to the point where freezing will occur at -20 degrees Celsius.