Der CS650 ist eine Multiparametersonde, die innovative Technik zur Bestimmung des volumetrischen Wassergehalts, der elektrischen Leitfähigkeit und der Temperatur nutzt. Er gibt ein SDI-12 Signal aus, das fast alle unsere Logger messen können.Lesen Sie mehr
The CS650 consists of two 30-cm-long stainless steel rods connected to a printed circuit board. The circuit board is encapsulated in epoxy and a shielded cable is attached to the circuit board for datalogger connection.
The CS650 measures propagation time, signal attenuation, and temperature. Dielectric permittivity, volumetric water content, and bulk electrical conductivity are then derived from these raw values.
Measured signal attenuation is used to correct for the loss effect on reflection detection and thus propagation time measurement. This loss-effect correction allows accurate water content measurements in soils with bulk EC ≤3 dS m-1 without performing a soil specific calibration.
Soil bulk electrical conductivity is also calculated from the attenuation measurement. A thermistor in thermal contact with a probe rod near the epoxy surface measures temperature. Horizontal installation of the sensor provides accurate soil temperature measurement at the same depth as the water content. Temperature measurement in other orientations will be that of the region near the rod entrance into the epoxy body.
|Sensing Volume||7800 cm3 (~7.5 cm radius around each probe rod and 4.5 cm beyond the end of the rods)|
Meets EN61326 requirements for protection against electrostatic discharge and surge.
|Sensor Output||SDI-12; serial RS-232|
|Warm-up Time||3 s|
|Measurement Time||3 ms to measure; 600 ms to complete SDI-12 command|
|Power Supply Requirements||6 to 18 Vdc (Must be able to supply 45 mA @ 12 Vdc.)|
|Maximum Cable Length||610 m (2000 ft) combined length for up to 25 sensors connected to the same datalogger control port|
|Rod Spacing||32 mm (1.3 in.)|
|Ingress Protection Rating||IP68|
|Rod Diameter||3.2 mm (0.13 in.)|
|Rod Length||300 mm (11.8 in.)|
|Probe Head Dimensions||85 x 63 x 18 mm (3.3 x 2.5 x 0.7 in.)|
|Cable Weight||35 g per m (0.38 oz per ft)|
|Probe Weight||280 g (9.9 oz) without cable|
|Active (3 ms)||
|Quiescent||135 µA typical (@ 12 Vdc)|
|Range for Solution EC||0 to 3 dS m-1|
|Range for Bulk EC||0 to 3 dS m-1|
|Accuracy||±(5% of reading + 0.05 dS m-1)|
|Precision||0.5% of BEC|
Relative Dielectric Permittivity
|Range||1 to 81|
Volumetric Water Content
|Range||0 to 100% (with M4 command)|
|Range||-50° to +70°C|
|Accuracy||±0.1°C (for typical soil temperatures [0 to 40°C] when probe body is buried in soil)|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
External RF sources can affect the probe’s operation. Therefore, the probe should be located away from significant sources of RF such as ac power lines and motors.
Multiple CS650 sensors can be installed within 4 inches of each other when using the standard datalogger SDI-12 “M” command. The SDI-12 “M” command allows only one probe to be enabled at a time.
The CS650G makes inserting soil-water sensors easier in dense or rocky soils. This tool can be hammered into the soil with force that might damage the sensor if the CS650G were not used. It makes pilot holes into which the rods of the sensors can then be inserted.
Number of FAQs related to CS650: 54
Alle anzeigenWenige anzeigen
Campbell Scientific strongly discourages shortening the sensor’s rods. The electronics in the sensor head have been optimized to work with the 30 cm long rods. Shortening these rods will change the period average. Consequently, the equations in the firmware will become invalid and give inaccurate readings.
Yes. There is surge protection built into the sensor electronics. The sensor survives a surge of 2 kV at 42 ohm line-to-ground on digital I/O and 2 kV at 12 ohm line-to-ground on power. It also survives a surge of 2 kV at 2 ohm line-to-ground on the rods.
If additional surge protection is required, consider using the SVP100 Surge Voltage Protector DIN Rail with Mounting Hardware.
The electrical conductivity (EC) of sea water is approximately 48 dS/m. The CS650 can measure permittivity in water with EC between 0 and 3 dS/m. EC readings become extremely unstable at conductivities higher than 3 dS/m and are reported as NAN or 9999999. Because EC is part of the permittivity equation, an EC reading of NAN leads to a permittivity reading of NAN as well. Thus, the CS650 cannot provide good readings in sea water.
With regard to sea ice, the electrical conductivity drops significantly when sea water freezes and the permittivity changes from approximately 88 down to approximately 4, as the water changes from a liquid to a solid state. With both EC and permittivity falling to levels that are within the CS650 measurement range, the sensor is expected to give valid readings in sea ice. The sensor is rugged and can withstand the cold temperatures. However, as the ice melts, there will be a point at which the electrical conductivity becomes too high to acquire a valid reading for either permittivity or electrical conductivity.
The CS650 and CS655 are warranted by Campbell Scientific to be free from defects in materials and workmanship under normal use and service for 12 months from the date of shipment. For further details, see the “Warranty” section of the CS650/CS655 manual.
The volumetric water content reading is the average water content over the length of the sensor’s rods.
CS650 and CS655 sensors are read one at a time using SDI-12 commands. Consequently, they are never active at the same time and do not interfere with each other electrically. When installing the sensors close together, a general guideline is to keep them at least 10 cm apart.
No. The principle that makes these sensors work is that liquid water has a dielectric permittivity of close to 80, while soil solid particles have a dielectric permittivity of approximately 3 to 6. When liquid water freezes, its dielectric permittivity drops to 3.8, essentially making it look like soil particles to the sensor. A CS650 or CS655 installed in soil that freezes would show a rapid decline in its volumetric water content reading with corresponding temperature readings that are below 0°C. As the soil freezes down below the measurement range of the sensor, the water content values would stop changing and remain steady for as long as the soil remains frozen.
The CS650 has rods that are 30 cm long, and the CS655 has rods that are 12 cm long. The difference in rod length causes some changes in specifications. For example, the CS650 is slightly more accurate in its permittivity and water content readings, but the CS655 works over a larger range of electrical conductivity. In addition, the CS650 handles a larger measurement volume and provides good accuracy in low EC (electrical conductivity) sand and sandy loam. The CS655 is typically more accurate in soil, works well over a wide range of soil textures and EC, and is easier to install because of its shorter rods.
In soil that is sandy, sandy loam, or loamy sand with low electrical conductivity, the CS650 is a suitable option because it has slightly better accuracy specifications than the CS655 and a larger measurement volume.