Das SRM-5A ist ein asynchrones Modem für lokale Kommunikation zwischen Datenlogger und Computer. Zwei SRM-5A Kurzstreckenmodems sind nötig, eins am Datenlogger und eins am Computer. Jedes sendet über ein vieradriges Telefonkabel oder CABLE2TB mit zwei verdrillten Leiterpaaren. Datenraten bis zu 9600 bps sind möglich.Lesen Sie mehr
Campbell Scientific recommends connecting the SRM-5A to the data logger's CS I/O port via the SC932A interface; a 9-to-25 pin adapter is required. It may be possible to connect the SRM-5A to the data logger's RS-232 port, but configuration is more complicated; contact Campbell Scientific for more information. Surge protectors on both ends of the transmission line are strongly recommended.
|Data Rates||Up to 9,600 bps|
|Transmission Mode||Asynchronous, full duplex, 4-wire operation|
|Transmission Range||Up to 12.2 km (7.6 mi) depending on data rate and wire gage|
|Line Interface||5-screw (4-wire and ground) connector block with cable strain relief inside plastic cover|
|Transmission Line||CABLE2TP (for short RS-232 or prewired enclosures) or unconditioned, 4-wire telephone line|
|Transmission Level||6 dBm|
|Terminal Interface||EIA RS-232-C/CCITT V.24, female integral 25-pin connector|
|Operating Temperature Range||0° to 60°C (Manufacturer's temperature specification is listed. Limited tests conducted by Campbell Scientific have shown the SRM-5A to perform at temperatures over -25° to +50°C.)|
|Altitude||Up to 3000 m (10,000 ft)|
|Humidity||Up to 95% (non-condensing)|
|Dimensions||5.2 x 5.2 x 1.8 cm (2.1 x 2.1 x. 0.7 in.)|
|Weight||38 g (1.3 oz)|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
|CR9000X||Although the CR9000X is compatible, the SRM-5A does not support the CR9000X's maximum communication rate and therefore may not be practical for many CR9000X applications.|
The SRM-5A is compatible with the now retired SRM-6A; the two can be used on either end of the same line.
Most manufacturers refer to cables compatible with this application as "PE39 Cable".
|Company||Part Number||AWG||Shield Type||Rodent Proof|
.005 Copper Clad Stainless Steel
|Company||Part Number||Outer Jacket||Messenger|
|Anixter||F-02P19AAF||19 HMW Polyethylene||.109 extra high strength steel|
Additional hardware is required to hang cables; contact your local Anixter representative for details.
Number of FAQs related to SRM-5A: 7
Alle anzeigenWenige anzeigen
Campbell Scientific does not currently offer any.
Technically, the SRM-5A is compatible with the CR200X-series dataloggers, but it is more complicated to use it with these data loggers. To use a SRM-5A with a CR200X-series datalogger, the SRM-5A must be in the DTE position on the station modem. In addition, user-supplied adapters and gender changers are needed to complete the connections between the SRM-5A and the RS-232 port of the data logger. Contact Campbell Scientific for more information.
Use the terminal mode of the Device Configuration Utility (select Unknown for Device type) set to 9600 baud. Click the Connect button to open the terminal screen.
On the SRM-5A, short together the RCV + to the XMT +, and the RCV – to the XMT – connections. Try typing letters and numbers with the computer keyboard. If these do not appear on the terminal screen, the modem is damaged and must be replaced. Ensure that the Echo Input box is NOT selected.
If connecting to the CS I/O port, one of the following is needed:
A connection can also be made to the RS-232 port. However, the switch on the SRM-5A must be in the DTE position on the station modem, and adapters and gender changers may be needed to complete the connection.
It is not cost effective to repair. The SRM-5A should be replaced.