Das NL115 von Campbell Scientific ermöglicht 10baseT Ethernetkommunikation mit dem Logger und die Erweiterung des Speicherplatzes über CF-Karten auf bis zu 16 GB. Es kann am Peripherieport eines CR1000 oder CR3000 angeschlossen werden.Lesen Sie mehr
The NL115 allows the datalogger to communicate over a local network or a dedicated Internet connection via TCP/IP. A 10baseT Ethernet straight through cable is used when the cable is run from a hub to the NL115. A 10baseT Ethernet crossover cable is used if the cable is run directly from the computer to the NL115.
The NL115/CF card combination can be used to expand the datalogger's memory, transport data/programs from the field site(s) to the office, and upload power up functions. One Type I or Type II CompactFlash (CF) card fits into the NL115’s card slot.
Only industrial-grade CF cards should be used with our products. Although consumer-grade cards cost less than industrial-grade cards, the consumer-grade cards are more susceptible to failure resulting in both the loss of the card and its stored data. Industrial-grade cards also function over wider temperature ranges and have longer life spans than consumer-grade cards.
Data stored on the card can be retrieved either by removing the card and carrying it to a computer or through a communications link with the datalogger. The computer can read the CF card either with the computer’s PCMCIA slot and the CF1 adapter or the computer’s USB port and the 17752 Reader/Writer.
CardConvert is used to convert and save binary data from a CompactFlash® card. CardConvert is included in our PC200W, PC400, LoggerNet, and RTDAQ datalogger support software.
|Power Requirements||12 V supplied through the data logger’s peripheral port|
|Ethernet Speed||10Base-T (10 Mbps), half-duplex|
|Typical Access Speed||200 to 400 kbits s-1|
|Memory Configuration||User selectable; ring (default) or fill-and-stop|
|CF Card Requirements||Industrial-grade|
|Data Logger Operating System (OS)||The CR1000 OS must be OS 9 or later. Both the CR1000 and CR3000 need OS 25 or later to read cards with more than 2 GB of storage.|
|Typical Current Drain||
|Weight||154 g (5.4 oz)|
EMI and ESD Protection
|European Standards||Meets requirements for a class A device under European Standards.|
|Application of Council Directive(s)||89/336/EEC as amended by 89/336/EEC and 93/68/EEC|
|Standards to Which Conformity Is Declared||EN55022-1; 1995 and EN50082-1: 1992|
|LoggerNet||Version 3.2 or later|
|PC400||Version 1.3 or later|
|DevConfig||Version 1.5 or later|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
|ENC12/14||Attached to a CR1000|
|ENC14/16||Attached to a CR1000 or CR3000|
|ENC16/18||Attached to a CR1000 or CR3000|
Campbell Scientific offers and recommends several CF cards (see Ordering tab). These cards have passed Campbell Scientific's ESD testing and operate properly with our data loggers.
Only industrial-grade CF card should be used with our products. Although consumer-grade cards cost less than industrial-grade cards, the consumer-grade cards are more susceptible to failure resulting in both the loss of the card and its stored data. Industrial-grade cards also function over wider temperature ranges and have longer life spans than consumer-grade cards.
The NL115 is set up using the Device Configuration utility (DevConfig). The NL115 requires LoggerNet 3.2 or later, PC400 1.3 or later, and DevConfig 1.5 or later.
The NL115 fastens onto the data logger's 40-pin peripheral port. This means that the NL115 is only compatible with the CR1000 and CR3000. It is NOT compatible with the CR800, CR850, CR200X, CR5000, or CR9000X dataloggers.
To read cards with more than 2 GB of storage, the data logger operating system must be OS 25 or later.
Number of FAQs related to NL115: 31
Alle anzeigenWenige anzeigen
No. The CR800 and CR850 do not have the necessary peripheral port to connect to an NL115.
The file on the card is open until the program is stopped or the button on the device is pressed.
Review the CRBasic Editor Help for information on the TableFile() instruction.
The best practice is to format the CF card, not using “Quick Format” on a PC, before inserting it into a device.
The NL115 isn’t configurable for an IP address. The IP address is set in the datalogger using the Device Configuration Utility.
If the CardOut() or TableFile() instructions are configured as fill and stop, the datalogger will stop writing data to the CompactFlash cards after they are full.
For detailed information, review the information in the “CF Card Information” application note.
The NL115 doesn’t have a MAC address. The MAC address is on the datalogger.
Yes. The data logger needs to be programmed to use FTP.